The Checker Plus electronic pH tester supplies you with fast and accurate readings on a large LCD. The electrode can be replaced, allowing the tester to last for years. Checker Plus is great for environmental, field and laboratory testing. Also suitable for checking electrodes in pipes and tanks and various laboratory applications. The Checker Plus pH tester includes a replaceable electrode with a screw-type connector, 2 sachets of 4.01 buffer solution, 2 sachets of 7.01 buffer solution, 2 sachets of cleaning solution and a plastic carrying case.
|Range||0.00 to 14.00 pH|
|Accuracy (@ 250°C/77°F)||± 0.2 pH|
|Calibration||Automatic, one or two point|
|Battery Type / Life||CR2032 Li-ion; 1000
hours, continuous use
|Auto off||8 min., 60 min. or can be shut off|
|Environment||0 to 50°C (32 to 122°F); RH 95% max|
|Dimensions (Meter only)||50 x 174 x 21 mm / 2 x 6.8 x 0.9"|
|Weight||50 g (1.85 oz)|
The pH electrode is shipped dry. Steps to take BEFORE using the meter:
Press the ON/OFF button to turn the meter on. All tags will be displayed. The meter will go into measurement mode: current reading and calibrated buffers are displayed.
In measurement mode, press and hold the ON/OFF button until "CAL" is displayed. When "7.01" blinks on the display, place the tip of the probe into a pH 4.01, 7.01, or 10.01 buffer solution.
Use "Procedure 1" for one or two-point calibration using pH 7.01 buffer.
Use "Procedure 2" for one-point calibration using pH 4.01 or pH 10.01.
For one or two-point calibration using pH 7.01 buffer
When pH 7.01 buffer solution is used as the first point, the buffer is recognized with the blinking stability indicator. When the reading is stable, the stability indicator will disappear and pH 7.01 will be calibrated.
When pH 7.01 is the only calibration point, finish this one-point procedure:
When using pH 4.01 or pH 10.01 as a second point, continue with the two-point procedure below:
For one-point calibration using pH 4.01 or pH 10.01 buffer
When pH 4.01 or pH 10.01 buffer solution is used as the first point, the value of the buffer is recognized and displayed with the blinking stability indicator.
|AUTO OFF||From measurement mode, press and hold the ON/OFF button. The meter will cycle through "OFF," "CAL," then current auto-off setting. The default setting is 8 minutes ("d08"). Press ON/OFF button to change default. "d60" is auto-off after 60 minutes. "d--" disables the auto-off feature. To Exit, Press and hold the button to exit the menu.|
|Clear Calibration CLr||Place meter in calibration mode. Press and hold ON/OFF until "CLr" is displayed. The meter will now be at default calibration. No tags will be shown in measurement mode until calibration is performed.|
|Err Message||In calibration mode, if the meter displays an "Err" message when in the correct fresh buffer solution then the probe should be cleaned. Place the probe in cleaning solution for 15 minutes. Rinse with purified water and place in storage solution for 1 hour before calibrating. If the "Err" message persists then the probe should be replaced.|
|Battery Indicator||When the battery is running low, the tag will blink on screen. When the battery has been depleted, "Erb" will appear on screen and the meter will turn off.|
Battery Replacement: To change the CR2032 Li-ion battery, turn the battery cover located on the back of the meter counterclockwise to unlock. Remove cover and replace with + side facing up.
Note: Batteries should only be replaced in a safe area using the battery type specified in this instruction manual. Old batteries should be disposed in accordance with local regulations.
Follow these tips to aid in attaining the highest accuracy for measurements
The EMS catalog number for the replaceable probe is 70700-01.